Blog post by Marco Altini
As previously reported we have added support for the CorSense sensor by Elite HRV.
CorSense is a sensor you can use rather than a cheststrap, and it is compatible with most Apple iOS and Android OS devices.
In this post, we'll show a few minutes of data collected under different conditions, highlighting how the sensor is very accurate in detecting RR intervals and can therefore be used reliably for HRV analysis.
Data was acquired using the CorSense sensor and a Polar H7 (previously validated with respect to ECG here), both connected to a different device running the HRV Logger app, which is an app that simply records everything coming from the sensor plus additional features.
During data acquisition, we collected data a few minutes while breathing freely, and a few minutes while deep breathing, to elicitate higher HRV due to RSA. You will see in the plots below visually the effect of deep breathing as we get greater swings in RR intervals.
A final note on data synchronization: data cannot be perfectly synchronized because it is not timestamped by the sensors. What we can do is either to log real time and then to split data in windows based on when data was collected, then compute HRV features on these windows or to sum up RR intervals over time. For this analysis we went with the second option and also tried to visually align the data streams.
We will start by looking at RR intervals, the basic unit we need to compute HRV features. RR intervals (peak to peak differences in consecutive heart beats) are provided by the two sensors directly, so we don't really need to do much to collect them, apart from linking the sensor to the HRV Logger app and export the csv files.
What can we derive from these data? You can see clearly almost perfect correlation between Polar H7 and CorSense for all conditions (relaxed vs paced breathing as highlighted by bigger oscillations in RR intervals or instantaneous heart rate), meaning that the sensor works really well in this modality.
Heart rate variability: rMSSD
As features, we will look only at rMSSD, the only feature we really care about. rMSSD is a clear marker of parasympathetic activity and the main feature we use for our analysis in HRV4Training, similarly to what other apps do as well. Additionally, the sports science community seems to have settled on this feature for several reasons (practical as well as it is easy to acquire, compute and reliable over short time windows and less controlled conditions), and therefore we'll stick to it.
What we expect given the data above is to see extremely close values between the Polar H7 chest strap and CorSense data.
For the plot below, I computed rMSSD for each time window (60 seconds in this case):
Results are very good considering normal variation in physiology and limitations in data synchronization.
Summary and other useful resources
That's all for this post. We are very pleased to see more and more sensors manufacturers spending time to work on 'HRV modalities' in which accurate RR intervals are sent via standard protocols, and hope that more will come in the future, making it easier for users to gather reliable data. Elite HRV did great work on the CorSense, as shown in the plots reported in this blog post.
Some additional resources:
Register to the mailing list
and try the HRV4Training app!
This blog is curated by
Marco Altini, founder of HRV4Training
The Ultimate Guide to HRV
1: Measurement setup
2: Interpreting your data
3: Case studies and practical examples
1. Intro to HRV
2. How to use HRV, the basics
3. HRV guided training
4. HRV and training load
5. HRV, strength & power
6. Overview in HRV4Training Pro
7. HRV in team sports
1. Context & Time of the Day
3. Paced breathing
4. Orthostatic Test
5. Slides HRV overview
6. Normal values and historical data
1a. Acute Changes in HRV
1b. Acute Changes in HRV (population level)
1c. Acute Changes in HRV & measurement consistency
1d. Acute Changes in HRV in endurance and power sports
2a. Interpreting HRV Trends
2b. HRV Baseline Trends & CV
3. Tags & Correlations
4. Ectopic beats & motion artifacts
5. HRV4Training Insights
6. HRV4Training & Sports Science
7. HRV & fitness / training load
8. HRV & performance
9. VO2max models
10. Repeated HRV measurements
11. VO2max and performance
12. HR, HRV and performance
13. Training intensity & performance
14. Publication: VO2max & running performance
15. Estimating running performance
16. Coefficient of Variation
17. More on CV and the big picture
18. Case study marathon training
19. Case study injury and lifestyle stress
20. HRV and menstrual cycle
21. Cardiac decoupling
22. FTP, lactate threshold, half and full marathon time estimates
23. Training Monotony
Camera & Sensors
1. ECG vs Polar & Mio Alpha
2a. Camera vs Polar
2b. Camera vs Polar iOS10
2c. iPhone 7+ vs Polar
2d. Comparison of PPG sensors
3. Camera measurement guidelines
4. Validation paper
5. Android camera vs Chest strap
6. Scosche Rhythm24
7. Apple Watch
9. Samsung Galaxy
1. Features and Recovery Points
2. Daily advice
3. HRV4Training insights
4. Sleep tracking
5. Training load analysis
6a. Integration with Strava
6b. Integration with TrainingPeaks
6c. Integration with SportTracks
6d. Integration with Genetrainer
6e. Integration with Apple Health
6f. Integration with Todays Plan
7. Acute HRV changes by sport
8. Remote tags in HRV4T Coach
9. VO2max Estimation
10. Acute stressors analysis
11. Training Polarization
12. Lactate Threshold Estimation
13. Functional Threshold Power(FTP) Estimation for cyclists
14. Aerobic Endurance analysis
15. Intervals Analysis
16. Training Planning
17. Integration with Oura
18. Aerobic efficiency and cardiac decoupling
1. HRV normal values
2. HRV normalization by HR
3. HRV 101